It is sued for micro-operation in dental especially, and can be used for micros-operation in ophthalmology, ear-nose-throat, gynecology and surgery. Microscope parameters: Dubleey epiece tube:Pupildistance adjustment rang :55mm-80mm Eyepiece:Field of vision adjustment :±7D Big objective lens:F=250mm Object diameter:LED light source>45,000LX View diameter:10mm-100mm Spot diameter:90mm Balance arm:Down240mm, Middle450mm, Up230mm Microscope light source:Balance the light source Maximum resolution:19201080P Camera shooting:Extraposition 3.35X22.5X Manual continuous regulation.
Camera parameters: Dubleey epiece tube:Dubleey epiece tube F=150mm, Pupildistance adjustment rang55mm-80mm Eyepiece10XField of vision adjustment±7D Bobjective lens F=250mm Object diameter LED light source >45,000LX View diameter:8mm-80mm Spot diameter:90mm Balance arm Down240mm; Middle450mm; Up230mm Photo light source:LED parallel light source>10000hours life Objective Magnification 3.35X22.5XManual continuous regulation Display method system :software display or IOS system software Maximum resolution:Full HD 19201080P Power supply: AC220V 50HZ/AC110V 60HZ Packing list: Microscope body1 Eyepiece2 Camera1 Adapter1 0.5 Object lens1 Handle2 Light Source1 Bracket1 Mechanical arm1 Fixture1 USB cable1 Power cord2 Cart1 Operation: DO NOT exposed to sunlight. Should be placed in a dry, clean environment, avoid high temperatures and severe vibration.The microscope is a precision instrument and should be handled carefully to avoid shocks and collisions during transportation. DO NOT leave dirt or fingerprints on the surface of the lens, so as not to affect the clarity of the image. Maintenance: All lenses should be kept clean. If there is any fine dust, blow it off with a blower or gently wipe off the cotton yarn.
If there are traces of oil and fingers, use a small amount of cotton yarn with a 3:7 ratio of ethyl alcohol and ether to gently wipe off. Do not use organic solvents to wipe the surface of the microscope, especially the surface of plastics, and use neutral detergents for cleaning. Do not disassemble the microscope by yourself so as not to affect the performance of the microscope Do not touch dust, cover it with a dust cover, store it in a place away from moisture, so as not to rust or mold, when the microscope is not in use. To maintain the performance of the microscope, regular inspections are recommended. The sale of this item may be subject to regulation by the U.Food and Drug Administration and state and local regulatory agencies. EUB (under 2KG): Only for USA, Canada, France, UK and Australia. China Post (under 2 KG): 25-55 days. E-Packet(under 6kg): Only for UK, Canada, TaiWan, HongKong, Singapore, Brazil(without bettery), Japan, Korea, Spain and Australia. Few states need to pay remote surcharge.
Don't worry, I will tell you before you pay if it's need. Please make sure it's correct. Your SATISFACTION is our ultimate goal. We work hard to earn your trust and future business. A Please check the package first before you sign for the package.
B If you have a defective item, you can return it or ask for a discount. Or replace item for you. We are factory founded dental equipment Wholesaler and Retailer.
If you have any questions about our service or products. You will be replied within 24 hours(48 hours in the weekend). The item "5W LED Operating Surgery Ophthalmic Microscope with Digital Camera Clip on Cart" is in sale since Monday, September 16, 2019. This item is in the category "Business, Office & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\Microscopes".royal" and is located in ShangHai. This item can be shipped to North, South, or Latin America, all countries in Europe, Japan, Australia.